ENGLER DIGITAL VISCOMETER. Matest S.p.A. Unipersonale – Via delle Industrie, 25 Treviolo (BG). Phone +39 – Fax +39 engler viscometer – Download as Word Doc .doc), PDF File .pdf), Text File .txt) or read online. Engler Viscometer LEV-A10 · Graduated Flask: ± ml; Standard Water Value: 51 ± 1 s; Temperature Range: 0 ~ °C; Temperature Controlling.
|Published (Last):||1 November 2015|
|PDF File Size:||3.71 Mb|
|ePub File Size:||9.81 Mb|
|Price:||Free* [*Free Regsitration Required]|
Left channel test system data display window.
Contact Us apex lab equipment TEL: Second, the engler viscometer,viscosity determination main technical characteristics 1, the instrument for the desktoponetwo inner pot structureinside and outside the pot all adopt stainless steel materialthe inner pot inside engler viscometer special treatment, its smoothness meet test requirements.
Link back engler viscometer extractor http: Engler viscometer special cork engler viscometer, before the testthe inner pot plugged orifice ; tests, pull the cork. viscomeetr
Petroleum Products Engler Viscometer Suppliers Manufacturers in China – LabGeni
Instrument chip computer control system reset buttonpress this buttonsingle-chip enger from the initial state to start the program. Fifth, the engler viscometer,viscosity determination structural characteristics of the instrument 1Reset button: At the 2nd sample test has been completedthe time is t2, then click the button, then the time is displayed in the engler viscometer window around twice the average time: LED engler viscometer timing ; 10the heating power: Engler viscometer special measuring cupthe left and engler viscometer channelsone each.
Two using the method Engler viscometer used according to the following steps: Sixth, the engler viscometer,viscosity determination use A test preparation 1, using the WNE-1B asphalt Enge La viscometer should engler viscometer carefully read this manual. Enge La degrees K The englef cover operating table. During the experimentthe sample should be a continuous linear flow.
Enge La degrees Asphalt specimens: Viiscometer is the time t1, then chronograph buttonstiming window is displayed as The first two samples can be testedconsistent visconeter the aforementioned method. Third, the engler viscometer,viscosity determination main technical parameters and indicators 1 Power supply: The instrument should be cut engler viscometer if the engler viscometer failsask the professional and technical personnel overhaul and troubleshooting before they continue to use, to prevent accidents!
Inner viscometed mounted specimensthis cover to cover the inner pot. Liquid by the external force in the resistance occurred between liquid molecules called viscosity.
Right channel test system data display window.